The JCMT relies on its in-cabin line-of-sight 183 GHz radiometer, the WVM (Water Vapour Monitor) to measure the PWV (Precipitable Water Vapour) and report the sky opacity at (the historically used) 225 GHz. The WVM is critical for both SCUBA-2 observations (see the note about extinction correction) and also for use by Nāmakanui, as well as determining the current weather conditions for programs to observe under, according to the JCMT’s Flexible Observing Guidelines.
The WVM works by obtaining measurements of the 183 GHz water line in a number of channels.
For archival values of the WVM you can visit: WVM Archive Interface.
Correcting WVM issues in SCUBA-2 data
If WVM data are missing from raw files, but the full data from the WVM are available on disk at CADC, the WESLEY recipe can be used to copy the WVM data into the raw files, and then reduce as normal. To obtain WVM data files visit first visit JCMT CADC archive page and search the JCMT’s collection using the selected instrument as: “wvm”. Then run up starlink’s ORACDR
wesley --files files.list INSERT_JCMT_WVM_DATA -verbose -log sf -nodisplay -recpars="JCMT_WVM_FILE=20200217.wvm"
ensuring 20200217.wvm is in cwd/ORAC_DATA_OUT and where files.list are the raw files you wish to reduce.
These are the information written into the WVM .wvm files:
1) isoTime Time and date string
2) airMass Air Mass
3) tAmb Ambient temperature (kelvin)
4) tSky Sky temperature 0 (kelvin)
5) tSky Sky temperature 1 (kelvin)
6) tSky Sky temperature 2 (kelvin)
7) tWat The effective temperature
8) tOff The line of site opacity
9) mmH2O_a The line-of sight water density in mm
10) mmH2O_z The water density at zenith in mm
11) finalTau The Tau at zenith
12) tSys The system temperature of receiver 0
13 tSys The system temperature of receiver 1
14) tSys The system temperature of receiver 2
15) azimuthDeg Antenna azimuth in degrees
Users with SCUBA-2 data taken 20210128
On UT 20210128 an adjustment was made to the WVM method that was in place for a day and impacts SCUBA-2 data taken on that date. Users that had data taken will need to follow the method outlines above, running the welsley command on raw SCUBA-2 data prior to reducing data. the .wvm file that should be used on that date is found at the following link: 20210128_OLD.wvm